Companion diagnostics (also known as CDx) are used together with a medicinal product. Physicians use them, for example, to make sure that a particular medicinal product is actually suitable for a patient. This means that CDx play a particularly big role in personalized medicine.
As the companion diagnostic and the medicinal product are inseparably linked to one another, there are some specific features in their development and authorization.
In this article you will learn:
The success of some drug therapies requires the use of an in vitro diagnostic device (IVD). Laboratories use this IVD to determine whether a biomarker that the drug targets is present in the patient. Physicians can use the analysis from the IVD to make sure that a particular therapy is actually suitable for their patient.
Usually, the CDx detects gene sequence changes or the presence of certain proteins in the analyte.
Typical platforms are:
If a particular result from the IVD is a prerequisite for a safe drug therapy, then this IVD is called a companion diagnostic (CDx). Ideally, CDx are developed in close collaboration with medicinal product manufacturers.
Based on Recital (11) of the IVDR, the main intended uses of CDx are as follows:
However, the definitions of CDx in Europe and the USA are different.
In Europe, the term “companion diagnostic” is defined by Regulation 2017/746 on In Vitro Diagnostic Medical Devices (IVDR):
Art. 2(7) IVDR
“‘companion diagnostic’ means a device which is essential for the safe and effective use of a corresponding medicinal product to:
(a) | identify, before and/or during treatment, patients who are most likely to benefit from the corresponding medicinal product; or
(b) | identify, before and/or during treatment, patients likely to be at increased risk of serious adverse reactions as a result of treatment with the corresponding medicinal product;”
Explicitly not CDx
Recital (12) IVDR
“Devices that are used with a view to monitoring treatment with a medicinal product in order to ensure that the concentration of relevant substances in the human body is within the therapeutic window.”
This exception is particularly important because it does not exist in the USA. The FDA's definition of companion diagnostic is different to the European definition:
The FDA defines “IVD companion diagnostic devices” as follows:
“An IVD companion diagnostic device could be essential for the safe and effective use of a corresponding therapeutic product to:
Identify patients who are most likely to benefit from the therapeutic product
Identify patients likely to be at increased risk for serious adverse reactions as a result of treatment with the therapeutic product
Monitor response to treatment with the therapeutic product for the purpose of adjusting treatment (e.g., schedule, dose, discontinuation) to achieve improved safety or effectiveness
Identify patients in the population for whom the therapeutic product has been adequately studied, and found safe and effective, i.e., there is insufficient information about the safety and effectiveness of the therapeutic product in any other population”
Manufacturers who want to place their in vitro diagnostic medical device for monitoring a treatment with a medicinal product on the market in Europe and the USA should make sure they are aware of these different definitions.
There is another group of IVDs that are not companion diagnostics but that could be confused with them: “complementary diagnostics.”
Complementary diagnostics are neither defined nor described in the IVDR. Their use is recommended in combination with a specific medicinal product, rather than being mandatory before the medicinal product can be used. Therefore, they are used only to help decision-making, not as a required part of the therapy.
This means complementary diagnostics do not have a close link with a drug in the way that companion diagnostics (CDx) do. As a result, from a regulatory perspective, complementary diagnostics are treated in the same way as “traditional IVDs,” i.e., the specific features that apply to CDx do not apply.
One FDA-approved complementary diagnostic is an immunohistochemical assay for the detection of PD-L1 proteins in patients with non-small cell lung cancer who are candidates for cancer immunotherapy (nivolumab).
This is a complementary diagnostic and not a CDx because a therapeutic benefit has been demonstrated for ALL patients who receive the medicinal product irrespective of their biomarker status. Nevertheless, there are some patients who respond better to the medicinal product than others: patients who express particularly high levels of the protein PD-L1.
Therefore, the use of the IVD is not mandatory, but does provide a prognosis for how effective the drug treatment might be.
CDx play a big role in “personalized medicine” where aim is to find a customized therapy for an individual patient or the causes of their disease and, if necessary, to continually adjust it. The classic use of CDx is in oncology. A look at the FDA's List of Cleared or Approved Companion Diagnostic Devices backs this up.
One of the first CDx was a system to detect the level of HER-2 (also known as ERBB2) expression in breast cancer. This test is based on an immunohistochemical procedure and is used to decide whether a patient is an eligible candidate for treatment with Herceptin (trastuzumab). Herceptin is authorized for the treatment of metastatic breast cancer but is not effective in HER-2 negative patients and may even have detrimental effects in these patients. Therefore, the use of a CDx is essential for identifying patients who may benefit from the therapy.
As CDx and medicinal products are interdependent, the development (and authorization) of the two runs in parallel. There are various phases in CDx development that correspond to medicinal product clinical phases. Figure 1 provides an overview. We explain the individual phases in more detail below.
As the purpose of the diagnostic and the drug are inseparably linked, the regulatory requirements for medicinal products and CDx also intertwine. The specific regulatory requirements for CDx take both the IVD (CDx) and the medicinal product into account.
The IVDR contains separate regulations for CDx. The following aspects are the most significant:
The specific regulatory features are:
All companion diagnostics are assigned to risk class C. However, there are additional regulatory requirements for CDx that go beyond the requirements for other class C IVDs (see below). There are examples in MDCG Document 2020-16.
Normally, the conformity assessment for class C devices is only carried out on one representative device per generic device group. However, it is different for CDx: Art. 48(7) of the IVDR states that each individual CDx device must go through the procedure. This means CDx manufacturers cannot take advantage of device groups.
CDx can be conformity assessed according to Annex IX of the IVDR. However, section 5.2 of the annex must also be taken into account.
It is particularly noteworthy that, in addition to the notified body, a medicinal products authority also has to be involved. This is usually the European Medicines Agency (EMA).
Please note: If the authority has queries, the deadline is extended by the amount of time needed to process the queries.
If a CDx is based on genetic testing, patient counseling, which is required for other genetic tests, is not required (Art. 4(3) IVDR).
Generally speaking: CDx always require clinical data and fall under the scope of the “certain performance studies” referred to Art. 58(2) IVDR.
The three pillars of clinical evidence – scientific validity, analytical performance and clinical performance – naturally also apply to CDx.
MedTech Europe has summarized further information on this topic.
Otherwise, there are no special requirements for CDx performance evaluations.
We have summarized information on the performance evaluation for you in a separate article.
The CDx's instructions for use must include the international non-proprietary name (INN) of the associated medicinal product that it is a companion test for.
Under German legislation, the CDx is not mentioned in the medicinal product's patient information leaflet (AMG Section 11).
The US authorities have more experience with CDx than the European authorities and notified bodies. The following specific regulatory features apply in the USA:
There is an FDA guidance document available on this topic: Developing and Labeling In vitro Companion Diagnostic Devices for a Specific Group of Oncology Therapeutic Products.
There is still a lack of guidance on CDx. However, we recommend reading the following documents that are available already:
Given the lack concrete European specifications, we recommend following the FDA's specifications. The information from MedTech Europe on the subject of clinical evidence is helpful. In addition, manufacturers should always be aware of the latest developments. They can use the Johner Institute's Regulatory Radar, for example, to make sure they stay up-to-date.
You can use the standard steps for authorization as a guide to safely launch a companion diagnostic on the market. However, note the specific requirements for CDx in each step.
Use the intended purpose of your in vitro diagnostic device to check whether it is actually a CDx. This is only the case if there is an actual dependence on the therapeutic agent (see section 1 “Definition”).
The next step is to set up your quality management system according to ISO 13485 and the IVDR. In this regard, there are NO differences from other types of IVD.
There are also no specific requirements compared to other IVDs with regard to the technical documentation according to Annexes II and III of the IVDR. The requirements of Annexes II and III of the IVDR must be met. See our comments on the performance evaluation (see the “Specific regulatory features” section).
IVD manufacturers should be in contact with their pharmaceutical partners during device development.
The conformity assessment procedure is more complex than with other IVDs (see the “Consultation with medicinal products authorities” section). It starts with the formal application to the notified body. In addition to the name and classification of the CDx, the specific intended purpose including the reference to the specific medicinal product, among other things, has to be specified.
The medicinal products authority, generally the EMA, is involved through the notified body. The EMA, notified body, IVD manufacturer and, if necessary, the pharmaceutical company as well should attend a pre-submission meeting intended to support the preparation of the application.
The consultation process follows the procedure described above (section 3 a).
To ensure that everything runs smoothly during the development and authorization of your companion diagnostic, we recommend the following:
Be discerning when choosing your partner from the IVD or pharmaceutical sector. Developing a CDx requires a huge amount of coordination and cooperation. For it to go smoothly, the framework conditions must be right. The CDx project will be easier if your partner has prior experience with companion diagnostics.
Functioning agreements and a high level of communication are important for guaranteeing the compatibility of the IVD and therapeutic agent.
The planning should be coordinated to ensure that IVD development is consistent with the requirements concerning the medicinal product.
Prototype development, including the analytical performance evaluation, should be coordinated to ensure that the clinical trial can be started on time and as planned.
The FDA recommends that:
“Although codevelopment as a process does not require simultaneous development of the IVD companion diagnostic and the therapeutic product from beginning to end, the availability of an IVD with “market-ready” analytical performance characteristics (i.e., a test that is completely specified with complete analytical validation and meets the therapeutic product sponsor’s expectations for performance) is highly recommended at the time of initiation of clinical trial(s) intended to support approval of the therapeutic product.”
The alignment of the therapeutic agent and CDx should be started as early as possible. It is therefore important for the medicinal product manufacturer to involve the IVD manufacturer as early as possible.
If a companion diagnostic accompanies a medicinal product, both development and authorization are more costly and complex than for a medicinal product or IVD on their own. The involvement of the medicinal products authority also adds time to the procedure. These factors should definitely be taken into account in your planning.
Companion diagnostics (CDx) are becoming increasingly important in modern medicine. They play a big role in personalized medicine in particular.
However, for manufacturers of IVDs or medicinal products, this means more work as CDx come with additional regulatory requirements. Good planning and effective coordination between all parties involved is therefore crucial. But if their development is successful, CDx can bring huge benefits and take therapy to a new level for patients.
The Johner Institute’s experts will be happy to help you with any questions you may have on IVD authorization strategies or on the issue of companion diagnostics (CDx) specifically. Simply contact us via our contact form or email us.